Projects per year
A limiting factor of tuneable diode laser spectroscopy (TDLS) with wavelength modulation spectroscopy (WMS) is the presence of background residual amplitude modulation (RAM) on the recovered 1st harmonic signal. The presence of this background term is due to direct modulation of the source laser power. This work presents a novel method to optically remove the unwanted background, with the major benefit being that measurement sensitivity can be increased. The recently developed phasor decomposition method1 (PDM), is a near IR (NIR) TDLS analysis technique that is used with the addition of the new RAM nulling method to recover gas absorption line-shapes. The PDM is a calibration free approach, which recovers the gas absorption line-shape and the isolated 1st derivative of the line-shape from the 1st harmonic signal. The work presented illustrates and validates the new RAM nulling procedure with measurements examining the 1650.96nm absorption line of methane (CH4) with comparisons to theory.
|Journal||Proceedings of SPIE: The International Society for Optical Engineering|
|Publication status||Published - Oct 2009|
- fiber optics
- tunable spectroscopy
- diode-laser spectroscopy
- RAM nulling
FingerprintDive into the research topics of 'Recent advance in tunable diode-laser spectroscopy with background RAM nulling for industrial applications'. Together they form a unique fingerprint.
- 1 Finished